Navigator Acquire Dialog
This dialog opens when the 'Navigator-Acquire at Points' menu item is selected and lets you control what actions are taken at a series of Navigator items marked as 'Acquire' or 'Tilt Series'. For items marked as 'Acquire', the primary actions available are to take and save an image, take an image and save it as a map, or run a script at each point. If there are only items marked as 'Acquire', the Tilt Series option is unavailable; if there are only items marked as 'Tilt Series', the other actions are unavailable. If there are both kinds of items marked, the dialog will open with the Tilt series option selected but the other options available; however, the selected action will be taken only on the appropriate kind of item ('Acquire' or 'Tilt Series') and items of the other kind will be skipped.
Prior to the primary action are a series of optional steps that can be run after moving the stage to each item's position. They are shown in the order in which they would be run:
Rough eucentricity - The rough eucentricity task should be selected if it is necessary that the specimen be eucentric and if the Z height might be off by more than 10 microns. It is run first because it does not retain alignment to a specimen position.
Autocenter beam - This task will condense the beam and take a picture to determine how to recenter it. Beam autocentering must already be set up for the current magnification in the Beam Autocentering Setup dialog box .
Realign to item - This task will realign to a selected specimen feature, but requires that the item be located within a sufficiently large medium-magnification map, as described in Realign to Item . If rough eucentricity is not being run during the acquire sequence, then the realign operation will be done at the Z height of the item being aligned to, since changing Z after aligning would lose the alignment. However, if rough eucentricity is being run, then the Z height found during that operation is retained for the realign operation. Thus, a new map acquired at this position will have the right Z height. There will be some error in realigning to the map because of the Z height difference. This error will be recorded for the new map and taken into account when realigning to this pair of maps in the future.
Cook specimen - This task will stabilize the specimen by pre-exposing it for a certain amount of time or to achieve a certain dose. The parameters for this operation must already be set up in the Specimen Cooking Setup dialog box .
Fine eucentricity - This task can be selected if eucentricity needs to be found with the highest accuracy, but the Z height should be within 10 microns of eucentric before running this task. If 'Realign to item' was also selected, the task is run in a way that maintains alignment to the specimen position (the Refine & Realign command).
Autofocus - Select this option to run the autofocus routine prior to the main action. If 'Only at start of group' is checked, then autofocus is run only when the first item in a group is encountered, or when an item is not a member of a group. If the
Run script - This task is available as a preliminary action for all kinds of primary tasks. When the primary task is tilt series acquisition, the script should be used for quick actions such as opening secondary files; if extended tasks are performed, the script is responsible for maintaining alignment to the specimen position. Even if the script is stopped or fails to complete successfully, the program will go on to the primary task, unless the script runs the command 'SkipAcquiringNavItem'. The program keeps track of the script choice separately for tilt series and other primary tasks.
Acquire map image - The image file must already be opened before starting the operation, unless a new file is set to be opened when the first item is acquired. As each area is acquired, it will be added as a map item with a label based on the label of the point being acquired (e.g., 12-A for point 12). The original point will no longer be marked for acquisition and it will no longer be drawn, to reduce clutter. Once the acquisition is finished, you could delete the original points.
Just acquire and save image - Again, the image file must be open before starting unless a file is set to be opened on the first item. This option is useful if you do not need to make maps of your acquired images.
Run script - Select the script number to run. When the script completes or stops with an error that cannot be resumed from, the Navigator will go on to the next item. If there is an error that can be resumed from, or if you stop the script with 'End' or 'Stop' and it is then resumable, then the Navigator will not go on to the next item and the status bar will show 'STOPPED NAV SCRIPT'. At this point you can do three things: 1) Resume the script, 2) abandon this script and go on to the next item by using 'End', or 3) terminate the acquisition by selecting 'End Acquire' in the Navigator menu. These 'End' or 'Stop' functions can be activated either from the Camera and Script control panel or from the Script menu.
Acquire tilt series - All parameters and file properties should already be set up when you go to start the series.
Restore scope state after aligning to item
This option is enabled when you select 'Realign to item' as an initial task and are running a script. If it is checked, then the current magnification and beam settings will be restored after running 'Realign to item'. Usually, this is appropriate, which is why it happens by default when making maps or taking images. However, if you are realigning to a high-magnification map embedded in a medium magnification map, then you could turn off this option if the beam and magnification were already carefully adjusted before acquiring the high-magnification map. That way, you would not have to readjust these settings before starting the acquisition, and you would be protected against the scope being in an undesirable state before starting a particular realign operation.
Close column/gun valves at end
If this option is available, you can set it to have the column or gun valve closed at the end of the set of acquisitions. On a JEOL without a FEG, this option can be used to turn the filament off instead.
Send email at end
If this option is enabled, you can select it to have an email sent to the address entered with Set Email Address in the Tilt Series menu when the acquisition finishes. An email will be sent if the acquisition stops with an error, but not if you stop it yourself. For this option to be enabled, an SMTP server and address to mail from must be specified in the SerialEM properties file. The program keeps track of this option setting separately for tilt series and other kinds of primary tasks.